Pulse amplitude range: 50-360V*
Max cable length between
computer and the box: 30m
Transducers Trigger input
Control signals/Power supply
* - software selected
Pulser circuit waveforms: The rising edge of the Trig In signal initiates the transducer charging process which takes about 3us. After this time a transistor switch which discharges the transducer is turned on (discharge time is about 20ns, but it can be longer, if the capacitance of the transducer is to large; the limit is reached, if the transducer is made from the standard ceramics, is 0.1 mm thick and has 8 mm diameter).
Because of the very low output impedance of the device (<1 Ω) and short discharge time the pulse generated with this
device could be concerned as a real pulse answer for the most transducers.
Copyright © 1989 - 2018 Optel. All rights reserved.
Limited, registered in the Companies Register by the regional court Wroclaw Fabryczna, VI Industrial Section of KRS, under the number 0000124439,
NIP: PL 8981047033, REGON 008375538, operating capital 364,500 PLN