Pulser: Step pulser, 50V -360V in 8 levels (the voltage depends on the used transducer), 20ns edge falling time, separate,
Pulser circuit waveforms: The rising edge of the Trigger signal (described as Trig Inp) initiates the transducer charging process which takes about 3us. After this time a transistor switch which discharges the transducer is turned on (discharge time is about 20ns, but it can be longer, if the capacitance of the transducer is too large; the limit is reached, if the transducer is made from the standard ceramics, is 0.1 mm thick and has 8 mm
Because of the very low output impedance of the device (<1 Ohm) and short discharge time the pulse generated with thisdevice could be concerned as a real pulse answer for the most transducers. Transducers with a parallel matching inductance should not be used, since they do not allow the transducer to be pre-charged (the inductance causes a short-circuit).
Copyright © 1989 - 2019 Optel. All rights reserved.
Limited, registered in the Companies Register by the regional court Wroclaw Fabryczna, VI Industrial Section of KRS, under the number 0000124439,
NIP: PL 8981047033, REGON 008375538, operating capital 364,500 PLN